HIOKI LUX HiTESTER 3423 Digital Light Meter Review
ToolBoom supermarket for engineers offers light meters of various brands. However, in order to help you learn more about lux meters, I have chosen exactly HIOKI LUX HiTESTER 3423 due to its wide measuring range.
Technical Specifications of HIOKI LUX HiTESTER 3423 Light Meter
|Display||LCD, 3 BBЅ digit (1999),
auto display indicator (AUTO),
hold display indicator (HOLD),
battery consumption display indicator, backlight, measuring unit (lx)
|Measurement scale interval||20, 200, 2 000 lx range: 1 count step
20 000 lx range: 10 count steps
200 000 lx range: 100 count steps
|Digital overflow||"OFF" is displayed|
|Measurement accuracy||± 4 % ± 1 dgt. (23 ± 5 ºC)|
|Analogue output||200 mV (DC voltage) full scale ± 2,5 % of full scale|
|Battery operating time||up to 25 hours|
|Effect of radiated radio frequency electromagnetic fields||± 120 digits at 3 V/m|
|Dimensions (W × L × T), mm||74 × 170 × 30
|Weight (including the battery), g||310|
|Package content||light sensor cap, 2 R6P batteries, carrying case, user manual|
|Accessories (not included in the package)||connection cable (RS 232, 9-pin), analogue output cable (3.5 mm connector)|
Silicon photodiode is built in the light meter. Such light sensor allows measuring illumination created by light sources with any spectral-response characteristics. Whereas accuracy of measurements constitutes 4% you can check compliance of premises or objects with illumination standards using HiTESTER 3423. Analogue output allows using light meter as sensor in automation systems.
Features of HIOKI LUX HiTESTER 3423 Lux Meter
The lux meter is extremely easy to use. There are three light meter control buttons placed on the panel.
Right lateral side of the digital lux meter contains power on/off button, power adapter jack (6 V, 100 mA) and analogue output terminal (3.5 mm).
So, let us deal with functions of HIOKI 3423 luxmeter in details:
- Change of the measurement range – "RANGE" button. There are five illumination measurement ranges in the digital light meter and due to this you can estimate value in any of them more precisely. "AUTO" shown on the display means that the device is using automatic mode of range selection. Luxes (lx) are used as measurement units. They are shown in the right bottom corner of the display. When the measured value exceeds the measurement range "O.F" indication is displayed.
- Holding readings on the display – "HOLD" button. You can fix readings of the digital lux meter shown on the display any moment. Such function is extremely useful if you have to hold luxmeter in the way when you do not see the
display. In such case you should wait 2-3 seconds until measurement is finished and press the "HOLD" button.
HIOKI LUX HiTESTER 3423 Lux Meter
- Display backlight – hold down "HOLD" button for more than 1 second. Backlight mode works only with "HOLD" function. Display backlight lasts for about 7 seconds. You can disable it by pressing "HOLD" button again.
- Performing zero adjustment – "0 ADJ" button. Indication "ADJ" and then "CAP" will be shown after pressing. When indication "CAP" is shown you should cover light sensor with the cap and wait a few seconds until display shows "0".
Design of HIOKI LUX HiTESTER 3423 Light Meter
HIOKI LUX HiTESTER 3423 is a compact, however, rather heavy luxmeter equipment with very durable body frame. Digital light meter package contains light sensor (silicon photoreceptor, photo detector) and counting unit (unit for device control and measurements processing).
You can connect them in two ways: attach light meter control unit directly to the sensor or use RS232 connection cable. Usage of connection cable will simplify measuring in hard-to-reach places.
You should use protection cap in order to avoid lightmeter damaging. The cap is the same as one used for performing measuring scale zero adjustment.
Power is supplied from two batteries. Light meter package also contains protective case which allows fixing it on the belt or hanging on the hand strap.
Application of HIOKI LUX HiTESTER 3423 Digital Lux Meter
I measured illumination of workplaces of my colleagues in order to test the digital luxmeter. The photometer is highly sensitive: any change of the position influenced readings of the lightmeter. Illumination of all workplaces by the daylight met the standards (300 lux). However, in the evening the digital light meter readings complied with the standard only in the places located directly under turned on fluorescent lamps. In such cases application of the digital photometer helps to detect whether illumination level is safe for eyesight.
In any industrial or residential premises illumination level must comply with the existing standards in order to provide comfortable environment for people to live and work, plants to grow, etc.
There are a lot of application fields of the digital lux meter:
- labour protection sphere;
- educational institutions and research centers, project organizations;
- museums, libraries, archives;
- transport and communication companies;
- standardization, metrology and certification centers;
- medical institutions;
- assessment of individual workplaces;
- agriculture, etc.
As I have already mentioned ToolBoom supermarket for engineers offers light meters of such brands: Uni-T, MASTECH, Pro'sKit and HIOKI. All of them have different price and quality of measurements. HIOKI LUX HiTSTER 3423 is significantly more expensive than others (for example, MASTECH MS6610 illumination meter). Difference in price is compensated by high quality of the light meter equipment, wider range of measurements and high accuracy of readings. Due to this you can use it to test premises for compliance with lighting standards.
Buy HIOKI LUX HiTESTER 3423 Light Meter (digital lux meter) in ToolBoom supermarket for engineers.
Technical specialist of ToolBoom online store